Heart valve assembly systems and methods

ABSTRACT

Heart valve assembly systems and methods configured for medical interventional procedures. In one aspect, the methods and systems involve a modular approach to treatment.

CROSS-REFERENCES TO RELATED APPLICATIONS

This application claims the benefit of U.S. Application Ser. No.61/635,741 filed Apr. 19, 2012 and U.S. Application Ser. No. 61/669,383filed Jul. 9, 2012, the entire disclosures of which are expresslyincorporated herein.

BACKGROUND OF THE DISCLOSURE

The present disclosure relates to medical interventional systems andmethods and more particularly, to artificial valve assembly systems andmethods. The long-term clinical effect of valve regurgitation is wellrecognized as a significant contributor to cardiovascular relatedmorbidity and mortality. In particular, there are two basicclassifications of mitral regurgitation (“MR”), primary and secondary.Primary MR results when there is either direct tissue pathology of thevalve structures or there is structural damage/alteration of one or morevalve structures (leaflets, chordae). Secondary MR is a functional lossof valve competence caused left ventricular dilatation, and secondaryalteration of mitral valve geometry following damage to the myocardiumand left ventricle (“LV”) causing. Whether valvular in origin leading toa ventricular problem or of ventricular/muscle origin leading to thevalvular problem, the effect of high levels of MR is significant oncardiopulmonary physiology, resulting in significantly elevated leftatrial pressures and pulmonary pressures, pulmonary congestion, andvolume and energy overload effects on the myocardium. This physiologycreates significant heart failure symptoms of shortness of breath anddecreased physical endurance, ultimately leading to death.

The decision to intervene on a regurgitant mitral valve relates to thelevel of mitral regurgitation, the symptoms of the patient as anindicator of progressive negative physiologic effect, and the functionalstatus of the left ventricle, specifically ejection fraction. The riskof intervention is weighed against the benefit of MR treatment.

The mitral valve is a therapeutic target of intervention/surgery earlyin the disease process of primary valvular disease because of MR'sdeleterious effects on heart/ventricular function if left untreated. Forpatients with moderate-severe or severe levels of MR combined with evena modest decrease in ejection fraction (“EF”), or the development ofsymptoms, surgical correction is indicated. In this situation, the riskof surgery in what is an otherwise healthy patient is far outweighed bythe beneficial effects of eliminating the long-term negative effects ofMR.

A more difficult question has been the patient with secondary orfunctional mitral regurgitation. In this situation, the patient haspre-existing LV dysfunction combined with heart failure symptoms, and adeveloping/worsening level of MR. The risks of intervention in thisscenario are much greater. The net benefit of surgically intervening toeliminate the MR has not been demonstrated. Symptomatic benefit has beenseen, but not a net mortality benefit. Therefore, it is usuallycontemplated or applied concomitantly when a patient is undergoingcoronary artery bypass graft CABG revascularization.

The classification of mitral regurgitation as primary or secondary is auseful to differentiate between the underlying disease processes thatled to the incompetent valve. These provide a starting point that candirect the type and timing of an intervention. However, it is notsufficient to fully describe the issues that direct a therapeuticapproach. Because the mitral valve is complex structurally,mechanically, and physiologically, a more detailed description andunderstanding of the abnormalities associated with mitral regurgitationis needed to direct existing therapies, as well as develop new optionsfor therapy.

Pathologic abnormality of the mitral valve tissue is a common cause ofprimary mitral regurgitation. Typical pathologies that occur includerheumatic, myxomatous, endocarditis, and Marfan's or other collagenbased tissue diseases. Calcification and leaflet thickening are alsoabnormalities associated with direct tissue level changes in the valve.These can be either part of a primary tissue based disease or resultfrom a long-standing insult to the valve, including regurgitant jettingacross the leaflets.

Congenital and acquired structural abnormalities like ruptured chordae,leaflet prolapse, fenestrations, and clefts can also be forms of primaryvalve disease leading to mitral regurgitation.

Functional MR results from myocardial damage leading to ventricularfunctional loss and geometric changes that impact the valve coaptationthrough associated annular dilatation and papillary muscle displacement.In pure functional MR, the valve structures are not pathologic nor havestructural defects, but the geometric alteration leads to a loss ofcoaptation of the mitral valve leaflets, often in the central A2/P2segment of the valve.

As with many multi-factorial clinical problems, one etiologic element(tissue pathology, structural alterations, functional/geometric changes)may lead to others resulting in a “mixed” picture. This is especiallytrue with mitral regurgitation. In the case of primary MR of eithertissue or structural origin, volume overload of the LV can createfailure and LV dilatation creating a component of functional MR if thevalve is left untreated. In the case of long standing functional MR,tissue changes can be seen such as calcification and thickening causedby the regurgitant jet and high leaflet stresses. Muscle/tissue damageto the myocardium in and around the sub-valvular apparatus can createstructural alteration such as ruptured papillary muscles/chordae andprolapse. Excessive tenting of the leaflets associated with significantfunctional MR can also stress the chords causing rupture.

The net result is that MR is a spectrum disorder with many patientshaving a mixed picture of valve abnormalities. This is an importantfactor in the decisions surrounding a mitral valve therapeutic approach,specifically repair or replacement.

The primary goal of any therapy of the mitral valve is to significantlyreduce or eliminate the regurgitation. By eliminating the regurgitation,the destructive volume overload effects on the left ventricle areattenuated. The volume overload of regurgitation relates to theexcessive kinetic energy required during isotonic contraction togenerate overall stroke volume in an attempt to maintain forward strokevolume and cardiac output. It also relates to the pressure potentialenergy dissipation of the leaking valve during the most energy-consumingportion of the cardiac cycle, isovolumic contraction. Additionally,successful MR reduction should have the effect of reducing the elevatedpressures in the left atrium and pulmonary vasculature reducingpulmonary edema (congestion) and shortness of breath symptomatology. Italso has a positive effect on the filling profile of the left ventricleand the restrictive LV physiology that can result with MR. Thesepathophysiologic issues indicate the potential benefits of MR therapy,but also indicates the complexity of the system and the need for atherapy to focus beyond the MR level or grade.

It is also desirable to prevent new deleterious physiology or functionof the valve. The procedure and system used to fix the mitral valveneeds to avoid worsening other (non-MR) existing pathologic conditionsor creating new pathologic conditions as a result of the treatment ofthe critical factors to be managed is Stenosis/gradient. That is, if avalve system is used that does not allow for sufficient LV inflowwithout elevated filling pressures, then critical benefits of MRreduction are dissipated or lost. Moreover, Atrial fibrillation is to beavoided as it can result if elevated pressures are not relieved by thetherapy, or are created by the system (high pressure results in atrialstress leading to dilatation ultimately leading to arrhythmias). Also,if the procedure results in damage to atrial tissue at surgery it canresult in the negative physiologic effect of atrial fibrillation.Further, one should be aware of the possibility of increased LV WallStress (LV geometry). Due to the integral relationship of the mitralvalve with LV geometry through the papillary and chordal apparatus, LVwall stress levels can be directly affected resulting in alterations ofLV filling and contraction mechanics. Accordingly, a system that doesnot preserve or worsens the geometry of the LV can counter the benefitsof MR reduction because of the alteration of contractile physiology.

It has been generally agreed that it is preferable if the valve can berepaired. Repair of valve elements that target the regurgitant jet onlyallows for minimal alteration to the valve elements/structures that areproperly functioning allowing for the least potential for negativelyeffecting the overall physiology while achieving the primary goal.Native valve preservation can be beneficial because a well repairedvalve is considered to have a better chance of having long standingdurability versus a replacement with an artificial valve that hasdurability limits. Also, while current surgical artificial valvesattempt chord sparing procedures, the LV geometric relationship may benegatively altered if not performed or performed poorly leading to anincrease in LV wall stress due to an increase in LV diameter. Thus,while preferred and possible for technically competent surgeons, therelatively high recurrence rate of MR due to inadequate repair, theinvasiveness of the surgery especially in sick or functional MRpatients, and the complexities of a repair for many surgeons lead to ahigh percentage of mitral operations being replacement.

Conventionally, surgical repair or replacement of the mitral valve isperformed on cardiopulmonary bypass and is usually performed via an openmedian sternotomy resulting in one of the most invasive high riskcardiac surgical operations performed, especially in subpopulations suchas functional MR. Therefore, a key improvement to mitral valveoperations is to significantly lower the risk and invasiveness,specifically utilizing a percutaneous or minimally invasive technique.

While there have been attempts to replicate existing surgical repair vialess invasive surgical or percutaneous methods, given the complexity ofrepairing the valve surgically, the efforts have largely been deemedlacking adequate efficacy and have not altered the risk benefit ratiosufficiently to warrant ongoing investment, approval, or adoption. Inparticular, there has been a general technology failure due to thecomplexity of anatomy to percutaneously manage with an implant orimplantable procedure. The broad spectrum of mitral disease directlyinfluences outcomes with a resulting inability to match technology withpathology. There has also been observed inadequate efficacy with poorsurgical replication and safety results. It has also been recognizedthat percutaneous approaches successful to certain valve procedures suchas aortic valve replacement associated with a single pathology and arelatively circular rigid substrate, mitral valves often suffer frommultiple pathologies and a flexible or elastic annular with multiplestructures.

Accordingly, what is needed is an effective long lasting MR reductionwithout creating negative physiologic consequences to thecardio-pulmonary system (heart, lungs, peripheral vasculature) includingstenosis, LV wall stress and atrial fibrillation. It is also desirableto be able to perform the operation in a reliable, repeatable, and easyto perform procedure and to have a broadly applicable procedure for bothpatients and physicians, while employing a significantly less invasivemethod.

The present disclosure addresses these and other needs.

SUMMARY

Briefly and in general terms, the present disclosure is directed towardsheart valve assembly systems and methods. In one particular aspect, thepresent disclosure presents various approaches to heart valve assemblysystems configured to eliminate MR, provide adequate physiologic inflow,and preserve and/or improve LV geometry.

In one aspect, there is provided a heart valve assembly system forimplantation at an interventional site including an anchor and a valveassembly defining structure configured to treat a native heart, and amethod for implanting the same.

In other aspects, there is provided a heart valve assembly system forimplantation at an interventional site which includes an anchor, theanchor including structure residing above and below a valve annulus andan interior and a valve assembly defining structure configured to beimplanted separately from the anchor and having an exterior sized andshaped to lockingly engage the interior of the anchor. In furtheraspects, the valve assembly can include a plurality of annular ridgesshaped to lockingly receive the anchor, or a generally cylindricalportion and a tapered portion extending from the generally cylindricalportion, the tapered portion contoured to mate with native valveanatomy. The valve assembly can additionally include a surfaceconfigured for tissue ingrowth, a projecting member extending beyond thetapered portion, the projecting member sized and shaped to engage nativevalve anatomy to offset rotational forces, and/or a tapered sectionextending within heart anatomy to engage a wall in a heart chamber.Moreover, the valve assembly can be configured to present structurewhich is at an angle with respect to a native valve opening, and definestructure supporting five or more leaflets. The valve assembly can alsoinclude leaflets which include non-coaptive tips which extend below thenative valve leaflets, a laterally arranged tri-leaflet arrangement,and/or leaflets which open and close in response to functioning ofnative leaflets, or which define a tubular structure with a wall thatcollapses to close the valve assembly. In yet another aspect, the valveassembly can include flap-like leaflets certain of which can be arrangedat different angles from other flap-like leaflets.

In various approaches, a heart valve assembly system and methodaddresses a number of basic functional requirements. One requirement isthe valve function itself, the occlusion of flow during systole, andopen to flow during diastole. Another requirement is the seal betweenthe artificial replacement valve frame/structure and the tissue toprevent/minimize any peri-valvular leaks or flow. A further requirementis the anchoring or securement function to hold the functioning valve inposition and withstand the substantial and variable cyclical load placedon the valve during systolic pressurization of the valve surface. It isintended that each of these is met in the durable, therapeutically, andphysiologically appropriate valve replacement system disclosed herein.

A valve replacement system according to the present disclosure includesa valve element and a valve delivery system. In accordance with thepresent teachings, the elements of the valve replacement system may beimplanted in staged procedures, for example, an anchor element may beimplanted during a first procedure and a valve element may be implantedduring a second procedure. As disclosed herein, the processes, systemsused for implantation, and timing of implantation may vary.

A valve to anchor interface can involve a geometric interlock, tothereby allow the flexibility for adaptation to a broad spectrum ofvalve technology. In this regard, a valve to native valve interfacepreserves sub-valvular structure relationships.

Further, design and delivery approaches that maintain native valvefunction providing the ability to completely separate and stage theimplantation of the system functional components is contemplated as aredelivery methods that have potential for quick fluoroscopic delivery,positioning, and deployment. Consequently, there is an optimal valveperformance opportunity due to maximal design flexibility and a deliverycapability to achieve precise positioning. The same creates desiredtissue/implant seating and maintains sub-valvular structuralrelationships.

Accordingly, employing the present system facilitates effective longlasting MR reduction without creating negative physiologic consequencesto the cardio-pulmonary system (heart, lungs, peripheral vasculature)including stenosis, LV wall stress, and atrial fibrillation. The systemfacilitates a reliable and a broadly applicable approach for bothpatients and physicians.

Other features and advantages of the present disclosure will becomeapparent from the following detailed description, taken in conjunctionwith the accompanying drawings, which illustrate, by way of example, theprinciples of the invention.

BRIEF DESCRIPTION OF THE DRAWINGS

FIGS. 1A and 1B are graphical representations, depicting characteristicsof potential patient populations;

FIG. 2A is a schematic drawing of the mitral valve anatomy at the levelof the mitral annulus;

FIG. 2B is a side view, depicting a portion of the schematic from FIG.2A;

FIG. 2C is a schematic section view of the mitral commissural area,showing the region of possible anchor and/or anchor projection tissueengagement;

FIG. 2D is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting possible locations for attachmentof the anchor to the valve tissue or anatomy;

FIG. 2E is a transverse (short axis) cross section of the heart at themitral valve annular level, depicting the commissural and posteriorleaflet cleft locations as possible attachment locations for the anchor;

FIG. 3 is a vertical cross-section of the heart, depicting the posteriorwall of LV with an exemplary anchor embodiment;

FIG. 4 is a transverse (short axis) cross section of the heart,depicting the mitral valve annular level of the exemplary embodiment ofFIG. 3, showing the circular anchor structure;

FIG. 5 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting the anchor of FIG. 3;

FIG. 6 is a cross-sectional view, depicting an exemplary embodiment of atri-leaflet artificial valve and anchor structure according to thepresent teachings;

FIG. 7 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting tri-leaflet deployable artificialvalve and exemplary anchor structure/substrate of FIG. 6;

FIG. 8 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting an alternative exemplaryembodiment of an artificial valve;

FIG. 9 is a top view of the valve of FIG. 8;

FIG. 10 is a vertical cross-section of the heart looking at theposterior wall of LV, depicting the valve of FIG. 8;

FIG. 11 is a view from the ventricular side of an exemplary embodimentof a five-leaflet mitral valve, depicting an envelope center leaflet;

FIG. 12 is a sectional view, the five-leaflet valve of FIG. 11;

FIG. 13 is a top view, depicting a multi-leaflet valve frame thatincludes a central triangular leaflet strut frame;

FIG. 14 is another embodiment of a multi-leaflet valve assembly usingthe frame of FIG. 13 in an open configuration;

FIG. 15 depicts a closed configuration of the valve assembly of FIG. 14;

FIG. 16 is a side view, depicting the valve assembly of FIG. 14;

FIG. 17 is a side view, depicting the valve assembly of FIG. 15;

FIG. 18 is a top view, depicting another embodiment of a multi-leafletvalve;

FIG. 19 is a sectional view of the valve assembly of FIG. 18;

FIG. 20 is a top view, depicting the multi-leaflet valve of FIG. 18;

FIG. 21 is a sectional view, depicting the valve assembly of FIG. 20;

FIG. 22 is a transverse sectional view, depicting an embodiment of afour-leaflet valve;

FIG. 23 is a sectional view of the left ventricle, depicting the valveof FIG. 22;

FIG. 24 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting a view of the valve in FIG. 22;

FIG. 25 is a vertical cross-section of the heart looking at theposterior wall of LV, depicting an embodiment utilizing dual replacementvalves;

FIG. 26 is a transverse section, depicting the dual valves of FIG. 25;

FIG. 27 is a vertical cross sectional view of through the aorta and theA2/P2 segment of the mitral valve, depicting the valve of FIGS. 25 and26;

FIG. 28 is a vertical cross-section of the heart looking at theposterior wall of LV, depicting an embodiment of a valve in a nativevalve;

FIG. 29 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting the valve of FIG. 28;

FIG. 30 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting the valve of FIG. 28;

FIG. 31 is a transverse top view at the mitral annulus level of FIG. 28;

FIG. 32 is a vertical cross-section of the heart looking at theposterior wall of LV, depicting an embodiment of a tubular valve;

FIG. 33 is a vertical cross section through the aorta and the A2/P2segment of the mitral valve, depicting the valve of FIG. 32;

FIG. 34 is a transverse cross section at the mitral annulus level with atop view of the valve of FIG. 32;

FIG. 35 is a top view, depicting an embodiment of a planar valveconcept;

FIG. 36 is a top view of the planar valve concept of FIG. 35;

FIG. 37 is a side view of the valve of FIG. 35; and

FIG. 38 is a side view of the valve of FIG. 36.

DETAILED DESCRIPTION OF THE PREFERRED EMBODIMENTS

Referring now to the drawings, which are provided by way of backgroundand example, and not limitation, the present disclosure relates tomedical interventional procedures and devices. In various aspects, heartvalve repair is addressed and in particular, mitral valve reductionapproaches are presented.

With reference to FIGS. 1A-B, there is shown a graphical representationof a potential patient population suffering from MR. Patents areclassified by valve abnormality versus the severity of symptoms (i.e.ejection retraction). A decision to be made involves whether to replaceor repair the subject valve. However, it has been found that a majorityof patients with MR are left untreated. This is especially true withfunctional MR. It has been determined that such patients can be treatedusing a percutaneous mitral valve implant approach.

In open surgical valve replacement, the valve is implanted in itsfunctional configuration and size. Additionally, conventional artificialsurgical valves have a sewing ring around their perimeter that isdirectly attached to the valve annulus tissue with multiple sutures toprovide both the securement and sealing functions. The surgical approachrequires the heart to be stopped (cardiopulmonary bypass) and the atriumto be opened.

For less invasive, beating heart approaches to valve replacement,whether trans-apical access or endovascular access (venous/antegrade,arterial/retrograde), the valve is not in a functional configuration andis in a compressed state to aid deployment. This requires the valve tobe deployed by some means to achieve its functional configuration andsize. The requirements of sealing and anchoring the valve must also havea deployment mechanisms and/or structures. These procedural operationsof deploying a functional valve, a tissue sealing structure, and a loadbearing anchor structure that is solidly secured and sealed to thenative anatomic location must be performed quickly and remotely toaccommodate the desired less invasive and beating heart implantation.This combination of multiple deployable elements with multiplefunctional requirements of the composite system dramatically increasesthe complexity of the system and procedure.

In general, the most difficult of the three functions to reliablyachieve can be the anchoring function due to the variable and cyclicalload requirements and the complexity of the anatomic structures of thenative mitral valve. The sealing function of the system is similarlydifficult because of the pressure requirements and again, the complexityof the anatomic structures of the native mitral valve. The simplest isthe deployable valve functional element, as the TAVI experience providesa basis for the starting point design structures and mechanisms.

It is desirable to have a simple and repeatable procedure to deliver ahighly functional and long lasting valve system requires a differentapproach than currently being pursued by others in the field.

In order to accomplish this, the presently disclosed system contemplatesa staged approach to the functional elements of the system, startingwith the anchoring or securement functional element. Additionally, thestaging can be performed within a single procedure or in multiple, timeseparated procedures. By staging and separating functional elements, theindividual elements will be simpler in design and simpler to deploy andimplant. This staging of the anchor implantation of the presentinvention provides a stable, reliable, consistent, substrate to delivera replacement valve into the mitral position.

In current conventional approaches to valvular intervention, adiagnostic echocardiograph is initially performed to assess valvefunction followed by two percutaneous valve procedures. First, adiagnostic angiography is performed with or without a right heartcatheterization to assess, for example, whether they might also requirerevascularization first, prior to intervention. Here, patients do notreceive valve therapy without the patient being fully revascularized.Thereafter, at a different time and place, valve replacement therapy isperformed involving fixation/attachment, accomplishing a tissue sealinginterface, and valve deployment and then release. In contrast, thepresently described approach, however, can include an assessmentinvolving a diagnostic echocardiography followed by a uniquepercutaneous valve procedure sequencing. First, a diagnostic angiography(+/−right heart cath) can be performed along with anchorfixation/attachment and anchor/tissue sealing. Subsequently, eitherlater or during the same interventional procedure, valve replacementtherapy can occur involving valve deployment and release. Thus, sincethe anchor implant allows the native valve to remain functional, theanchor implantation procedure could be added to the end of the angio(+/−PCI) and not require a separate interventional procedure. A quick,simple, and reliable anchor deployment could permit a fully ingrownstructure that significantly enhances the holding force of asubsequently implanted replacement valve. Tissue ingrowth of the entireanchor perimeter or at key positions thereon can in fact provide thenecessary tissue seal in advance of valve deployment. Moreover, theanchor design could be simplified due to less required acute holdingforce. Therefore, a tissue incorporated and healed anchor provides astructure to perform several methods of annular adjustment, includingplication, reduction annuloplasty, and septal-lateral cinching.

There are certain desirable anchoring locations for an anchor implant.Direct attachment to tissue is contemplated at locations adjacent themitral valve, as are locations for placement of anchor projections atleaflet cleft locations. Again, it is intended that there be low or noimpact to native leaflet function as a result of the implantation of ananchor implant so as to maintain the pre-existing native valve functionuntil a replacement valve is implanted. At the mitral valve 50 (SeeFIGS. 2A-2E), there is of course the mitral annulus 52 definingstructure from which the anterior leaflet 54 and posterior leaflet 56extend and articulate. Between the anterior and posterior leaflets 54,56 are commissural leaflets 58. The trigones 60 are positioned at aperimeter of the anterior leaflet 54 and adjacent the commissuralleaflet 58. Commissures 62 are the openings or slits dividing theanterior leaflet 54 form the commissural leaflets, and positioned nearthe trigones 60. Such structure defines consistent and predictableanatomical features across patients. Notably, the high collagen annulartrigone 60 generally can be relied upon to present a strong anchoringlocation. The muscle tissue in this area also provides a good ingrowthsubstrate for added stability. There is also a potential for sub-leafletattachment for more stability (See FIG. 2C). Accordingly, primaryanchoring locations 62, 64 for an anchor implant are included in FIGS.2D and 2E.

Turning now to FIGS. 3-5, there is shown one embodiment of an anchorimplant 100 configured for atrial anchoring and implantation within theheart 102 at the mitral valve annulus 104. The anchor implant defines asupra-annular ring sized and shaped to be placed at the annulus, andincludes commissural projections 106. As shown in FIG. 3, theprojections 106 can be placed at an anterior commissural trigone 108. Asdescribed above, the commissural projections 106 are configured toextend between leaflets 109 without interfering with their functions(See FIG. 4). Moreover, as shown, the implant 100 includes a generallycircular body 110 which can be formed from a wire or other structure,and the projections 106 are loops extending away from a plane defined bythe circular body 110. It is to be further recognized that the body 110includes a pair of bends 112 configured on opposite sides of theprojections 106 to thereby provide necessary stress relief and clearancefor the placement of the projections between leaflets 109. Furthermoreas noted previously, the anchor 100 can be covered with variousmaterials, such as PET and ePTFE, so as to present a desiredbiocompatible surface to body tissue.

Once the anchor is placed at a native valve, an artificial heart valvecan then be implanted. Alternatively, it is contemplated that theartificial heart valve assembly can be implanted without the aid of ananchor, but rather includes its own anchoring substructure intended tosecure the assembly in place.

One intention of mitral valve replacement is to maximize valve inflowarea by implanting as large a valve as feasible to avoid stenosis orsignificant inflow gradients and to prevent the pulmonary andright-sided negative consequences of elevated left atrial pressures andpossible pulmonary hypertension. Another consideration is to be able toimplant a valve best suited for a particular patient (age, clinicalstatus, etc.) or a particular valve pathology (functional MR,structural, mixed) without compromising the decision on valve size ortype (tissue, mechanical). Another goal for percutaneous implantation isto be able to leverage the prior developments utilized in TAVI,specifically having a system that allows for both balloon expandable andself-expansion deployment mechanisms. When desired, the staging andseparation of the implantation of a predictable and reliable anchorsubstrate disclosed herein provides for multiple options for anartificial valve structure. It is contemplated that the valve mayinclude a structural frame or support component, a leaflet/occludercomponent, and an attachment feature or component.

In one approach, the valve structure may comprise a single valve. Astable and predictable, relatively circular and non-expandable anchorstructure as disclosed herein is that it can secure a deployed circularvalve without requiring a frictional fit between the anchor and valve,which can reduce the required radial hoop strength (collapse force)requirements of the valve frame. The interlock fit rather than anexpanded frictional fit reduces the structural requirements of thevalve, allowing the structural design to focus on valve performanceloads rather than anchoring loads. The anchor structure may also allowfor a larger surface area valve because of the separation of functions.If a single valve is utilized, it may consist of a traditionaltri-leaflet construction with sufficient size (cross sectional area) tominimize flow restriction. To further maximize area and minimize leafletstresses, more than 3 leaflets can be utilized.

In another approach, the valve structure may comprise dual or multiplevalves. Implantation of dual parallel valves in the mitral orifice is anoption as a means to achieve the desired cross sectional area of thevalve while minimizing the loads and stresses of the individualartificial valves due to their smaller size. The substrate structure toreceive the dual smaller circular artificial valves can be achievedeither through an anchor that has a dual orifice planar configuration orvia a separate interface implant that connects to a larger circularanchor ring but also has the dual orifice substrate.

In yet another approach, the valve structure may comprise dual seriesvalves. The nature of a serial valve above the native allows for areduction in impulse hemodynamic load of the artificial valve becausethe native valve is absorbing most of the immediate systolic forces. Theartificial serial valve must be able to withstand the pressure timessurface area forces during mid to late systole but does not need towithstand the immediate impulse force. This is important because it mayallow for a reduction in the immediate anchoring load requirements ofthe system. This type of configuration also may allow for occlusion ofthe only the regurgitant volume of the native valve rather than theentire flow across the mitral orifice.

Moreover, in yet another aspect, the valve structure may comprise avalve within a valve. The nature of a valve in a valve is that theartificial valve only covers a fraction of the total valve area andtherefore only encounters a fraction of the total anchoring forcesrequired. Further, occluder designs and mechanisms may be utilized toprovide valve function. These types of designs are constructed primarilyfrom tissue or polymer constructs and rely on the valves geometryrelative to the pressure and flow of the blood to occlude backward flowand distribute the load on the valve surfaces over a larger portion ofthe tissue.

One ability of the contemplated approaches is to implant a mechanicalvalve in a beating, functioning heart via a less invasive trans-apicalor transatrial approach. Here, the mechanical valve can be attached orsecured to anchor structure in a staged approach. During the procedure,the mechanical valve is attached around its annular perimeter to anintroducer tool such that it can be tilted and rotated by the introducerduring insertion and implantation. When tilted, the valve annularperimeter is in plane with the long axis of the introducer tool. Thevalve is inserted into the heart via a slit created in the atrium or LVwall of the heart. In order to create the slit while maintaining heartfunction, a portal device can be used and is constructed of a proximalhub/port, a collapsible oval or slit like tubular segment, and a distalsegment that is attached to the heart via a purse string suturearrangement around its distal perimeter. This portal device serves as avestibule like structure for sealed introduction of the valve andinsertion tool. The distal end of the portal device is first attached tothe atrial or ventricular wall via purse string sutures around theperimeter to create a blood tight seal. A slit is then created in theheart wall from within the portal device with the proximal hubpreventing leakage or bleeding. The distal end of the tubular segment isthen clamped while the valve and introducer distal end is inserted intothe proximal segment of the portal tubular segment. The proximalhub/port is sealed around the introducer, air is evacuated and thedistal clamp is then released allowing advancement of the flat planarvalve and introducer structure through heart wall slit into the chamber.The valve is then further advanced, tilted, and rotated into positionacross the mitral valve and an engagement mechanism around the valveperimeter is used to engage and attach the mechanical valve to theanchor structure. Because a mechanical valve can be functional duringdeployment, no rapid pacing is anticipated during deployment. Aftervalve deployment and insertion tool removal, the purse string suturesare tightened to seal the heart wall.

As stated, in general, the valve can consist of a support structurecomponent and an occluder/leaflet component. The support structure canbe the component or portion of the artificial valve that transfers loadsfrom the occluder/leaflets to the anchor and/or tissue within theimplant site. The support structure functions to hold theoccluder/leaflets, transfer load into frame, and incorporates attachmentcomponents to attach to a previously placed anchor and/or tissue. Theleaflet support component of the support structure attaches directly tothe leaflets (sutures or alike) and supports and maintains the leafletgeometry as the valve opens and closes. The leaflet support componentcan be attached or be integral to the rest of the support structure. Theattachment component of the support structure consists of a mechanismthat allows stable and durable attachment of the support structure tothe anchor. The attachment component interfaces with and transfers loadto the anchor. There are several types of attachment methodscontemplated. In one approach, there can be discrete connection pointswith mechanical fasteners consisting of small interlocking components.There can also be attachment components that provide broad geometricinterlock between the support structure and anchor. These create ageometry that engages the anchor to provide a stable and durableattachment. Direct tissue attachment components could also beincorporated into the attachment component of the support structure.They would include barbs, hooks, as well as chronic ingrowth componentssuch as mesh, porous, and velour materials. The support structure hassimilar requirements to other implanted components such as the anchor.For example, the support structure can be with tissue and blood,biostable (does not significantly degrade) corrosion resistant(resistant to general corrosion as well as fretting, galvanic, pitting,crevice, and stress corrosion), and durable/fatigue/resistant (functionunder the in vivo cyclically load conditions for the implant lifetime).Additionally, the support structure must be of sufficient mechanicalstrength to transfer the loads from the occluder/leaflets to the anchor,and be small enough to allow delivery though and catheter and minimizethe amount of obstruction to blood flow after implantation.

As stated, the occluder/leaflet component is the component that preventsor inhibits backflow of blood from the left ventricle to the leftatrium. One design requirement of an occluder/leaflet is durability.This is especially the case in the mitral position because of the hightransvalvular pressure differentials. The leafletstake the load producedby the blood pressure gradient across the valve and transfers it to thesupport structure. Leaflets undergo large cycle displacements andtherefore bend and fold repetitively. Occluder/leaftlets alsorepetitively interact with other valve structures, primary with otherleaflets to create a seal (coaptation). This interaction creates wearand abrasion conditions which need to be addressed in the valve design.In order to enhance the durability of occluder/leaflet, minimizingstress within the leaflet, motion/strain of the leaflet and motion atcontact areas with other structures (e.g. coaptation area) is required.Material selection can be critical to the durability and the function(flexibility). Geometry of the leaflets can minimize high local stressesand motion. Three leaflet configurations provide convenient geometricrelationships in a circular configuration that help manage durabilityissues (stress and wear). Multiple leaflets provided smaller surfacearea per leaflet to occlude the same area. This reduces the stress andmotion within an individual leaflet. Multiple independent valves can beused to accomplish the same objective.

Materials used to build the valve depend upon the intended valvefunction and structure. Metallic, tissue, and synthetic materials andcombinations thereof may be used for the valves. Thus, metals may beused for the valve frame structures and can include Nitinol due to itssuperelasticity and ability to be compressed into a deliverableshape/state and then deployed into a functional state. Titanium can alsobe employed due to its strength and biocompatibility, or SST which ishardened for its strength or malleable to aid in conforming to shape.Moreover, a cobalt/chromium alloy is suitable for strength and a knownvalve component implant history, or composites can be used to providemultiple properties based on anatomic location. Further, tissue may beused for the occluder including pericardial (bovine, ovine, porcine)tissue or valve tissue (bovine, ovine, porcine), and synthetic polymerscan be used as biocompatible elements in implants. In this regard,Elast-Eon (a silicone and urethane copolymer), ePTFE, Urethane,Silicone, PEEK, and/or Polyester (PET) can be used as well as UHMWP.

Turning now to the figures, there are presented various contemplatedapproaches to valve assemblies. Although described in context with themitral valve, such structure can also be adapted for other heart valves.In one approach (FIGS. 6-7), a valve assembly 340 sized and shaped toengage anatomy at a native mitral valve can include a geometric lockingstructure 342 for facilitating a secure implantation at a mitral valveannulus. It is intended that the valve to native valve interfacepreserves subvalvular structural relationships. Thus, the device can beprovided with annular ridges 344 sized and shaped to lockingly receivean anchor implant 345 such as that described above and which waspreviously placed within the interventional site. Moreover, it is to berecognized that valve assemblies disclosed herein can be configured toengage one or more of the anchor structures depicted in copending U.S.application Ser. No. 13/842,206, the contents of which are incorporatedherein by reference. A downstream portion of the valve assembly 340 caninclude a tapered portion 346 contoured to mate with native mitral valveanatomy. The tapered portion 346 can be further equipped with a tissueengaging or ingrowth surface such as a woven structure which extendsfrom a generally cylindrical upper section 347. A downwardly projectingmember 348 is further provided to stabilize the valve assembly 340 inplace against rotational forces.

As shown in FIGS. 8-10, it is also contemplated that a valve assembly350 can include structure adapted for placing the assembly at an angleacross a native valve opening to be better responsive to pressure loadswithin the heart. Here, a loading vector with a posterior wall of theheart and anterior annulus anchor 351 can be improved by providing thevalve assembly with a tapered section 352 which extends within andengages an internal wall of the LV. Thus, at its upper end, the valveassembly 350 securely engages the anchor implant 351 and at its lowerextremity, the tapered portion 352 is supported by the LV, either withacute fixation elements such as anchors into the LV wall, or with tissueingrowth into the tapered portion 352, or both. With this arrangement,the valve assembly presents an angled profile suited to be responsive toloading within the native valve structure. In one approach, valveassembly 350 leaflets are angled with respect to native valve leaflets.

It has further been contemplated that various different multi-leafletvalve approaches may be best suited for particular patients orapplications. With reference to FIGS. 11 and 12, a five leaflet valveassembly 360 is presented. An envelope center valve 362 is bounded byfour separate peripheral 120° degree valve leaflets 364. The envelopecenter valve 362 is attached at opposite ends to an outer valve frame366 and folds along a center line. The center valve outer edges meet theedges of the peripheral leaflets 364 to close the valve assembly.Adjacent peripheral leaflets also engage each other when the valveassembly 360 is closed.

In FIG. 13, there is shown a valve assembly 370 including amulti-leaflet valve frame including a central three piece crossing strutframe 372 from which individual leaflets can articulate. In one approachto a valve assembly employing the strut frame (FIGS. 14-17), a fabriccovered valve assembly 380 can include a triangular coaptation zone 382.The central strut frame 384 supports three centrally located leaflets385 which are surrounded by a generally B-shaped second frame 386, thesame supporting an additional three outer leaflets 386, inner borders ofwhich being configured to meet outer borders of the central leaflets 385to define the triangular coaptation zone 382. It is noted that in thisparticular approach, two of the outer leaflets have a generally mirrorimage, while the third outer leaflet has a much narrower profile.

In yet another valve assembly 390 including a central strut frame 392(FIGS. 18-21), there are four leaflets, a single internal leaflet 394supported by the center frame 392, bounded by three external leaflets396 having general sizes and shapes. A second frame 397 which engagesthe internal frame has a generally circular shape. Coaptation occursbetween external edges of the internal leaflet 394 which folds intothree sections when opened, with the internal edges of the externalleaflets 396. Along an outer section of each arm of the internal centerframe 392 towards the outer frame 397, portions of the adjacent outerleaflets 396 engage in a coaptation.

With reference to FIGS. 22-24, there is shown a valve assembly 400having four heavily redundant leaflets 402. This assembly furtherincludes a commissural anchor structure having feet projections 404extending down into the LV, where non-coaptive tips 406 of the leafletsattach. Such attachment within the LV is beyond and/or displaced fromnative valve substructure.

In an alternative approach (FIGS. 25-27), a dual valve assembly 410 iscontemplated. Accordingly, the assembly 410 can include two tri-leafletvalves 412 arranged side-by-side within a generally oval shaped skirtframe 414 covered with a wire mesh fabric. The frame 414 is sized andshaped to securely engage or include anchors 415 configured throughvalve commissures. A pair of adjacent internal circular frames 416 arefurther provided to support the tri-leaflet valves 412.

A valve assembly can alternatively include structure that is responsiveto action of the native valve. As shown in FIGS. 28-31, a valve assembly420 suspended within the coaptation line of a native valve can include agenerally oval, narrow frame 422 sized to extend across a native valveopening. Opposite long ends 424 can be configured to engage an anchorimplant 425 placed at the native valve, or can include projectionsengaging the LV walls itself. Longitudinally extending valve leaflets426 are open (FIG. 29) when the native leaflets are open, and close(FIG. 31) in response to the closure of native leaflets. The valve isintended to be anchored in valve commissures and/or to the trigone.

As shown in FIGS. 32-34, a valve implant 430 can also embody a tubularbody structure 432. A ring-like frame 434 defines an opening to thevalve 430, and extending downwardly from the frame is the tubular body432. A lateral section 436 of the tubular body 432 can be configured tosupportively engage or be mated to the LV wall. The assembly alsoincludes an opposite lateral wall 438 defining collapsible structurewhich is closeable to prevent flow.

A valve assembly 440 can further include a frame 442 supporting aplurality of multiple slot or flap-like leaflets 444 formed in a surfaceextending across the frame 447 (See FIGS. 35-38). Each of the leafletscan be arranged in similar directions as shown in the figures, or can bearranged to present varyingly arranged openings, to thereby open andclose to control blood flow.

Further modifications and alternative embodiments will be apparent tothose of ordinary skill in the art in view of the disclosure herein. Forexample, the systems and the methods may include additional componentsor steps that were omitted from the diagrams and description for clarityof operation. Moreover, those of ordinary skill in the art willappreciate that aspects and/or features disclosed with respect to oneembodiment in some case may be incorporated in other embodiments even ifnot specifically described with respect to such other embodiments. It isto be understood that the various embodiments shown and described hereinare to be taken as exemplary. Elements and materials, and arrangementsof those elements and materials, may be substituted for thoseillustrated and described herein, parts and processes may be reversed,and certain features of the present teachings may be utilizedindependently, all as would be apparent to one skilled in the art afterhaving the benefit of the description herein. Changes may be made in theelements described herein without departing from the spirit and scope ofthe present teachings and following claims. Accordingly, thisdescription is to be construed as illustrative only and is for thepurpose of enabling those skilled in the art the general manner ofcarrying out the present teachings. It is to be understood that theparticular examples and embodiments set forth herein are nonlimiting,and modifications to structure, dimensions, materials, and methodologiesmay be made without departing from the scope of the present teachings.Other embodiments in accordance with the present disclosure will beapparent to those skilled in the art from consideration of thespecification and practice of the invention disclosed herein. It isintended that the specification and examples be considered as exemplaryonly, with a true scope and spirit being indicated by the followingclaims.

Thus, it will be apparent from the foregoing that, while particularforms of the invention have been illustrated and described, variousmodifications can be made without parting from the spirit and scope ofthe invention.

We claim:
 1. A mitral heart valve assembly system configured for stagedimplantation at a native mitral valve site via an endovascular access toa heart, comprising: an anchor to implant along a native mitral valveannulus while the native mitral valve remains functional, the anchorincluding: a supra-annular structure configured to reside above thevalve annulus of the native mitral valve and defining a first geometricinterlock face at an uppermost portion of the anchor; and a plurality ofcommissural projections that extend downwardly away from thesupra-annular structure and that are sized and shaped to extend betweenthe native mitral valve leaflets without interfering with thefunctionality of native mitral valve leaflets, wherein the commissuralprojections are configured to anchor along native heart tissue below thevalve annulus of the native mitral valve without interfering with thefunctionality of native mitral valve leaflets; and a valve assemblyconfigured to subsequently implant after the anchor in the stagedimplantation via the endovascular access to the heart, wherein the valveassembly includes: an occluder component having a plurality of leafletsformed from pericardial tissue and positioned to occlude blood flowduring systole and open to blood flow during diastole; a supportcomponent to maintain a portion of the occluder component in asupra-annular position above the valve annulus of the native mitralvalve and further having a second geometric interlock face sized andshaped to lockingly and directly engage the first geometric interlockface of the supra-annular structure of the anchor; and a lower contouredportion configured to engage with the native mitral valve, wherein atleast a portion of the lower contoured portion is tapered inwardly andconfigured to extend below the valve annulus of the native mitral valve.2. The system of claim 1, the valve assembly further comprising aplurality of annular ridges on said support component shaped tolockingly receive the anchor.
 3. The system of claim 1, the valveassembly further comprising a generally cylindrical portion and thelower contoured portion extending from the generally cylindricalportion.
 4. The system of claim 3, the valve assembly further comprisinga surface configured for tissue ingrowth.
 5. The system of claim 4, thevalve assembly further comprising a projecting member extending beyondthe lower contoured portion, the projecting member sized and shaped toengage native valve anatomy to offset rotational forces.
 6. The systemof claim 1, wherein the valve assembly is configured to present valveleaflet structure which is at an angle with respect to a native valveopening.
 7. The system of claim 1, wherein said plurality of leafletscomprises five or more leaflets.
 8. The system of claim 7, wherein oneleaflet of the five or more leaflets is an envelope center valve boundedby four separate and distinct leaflets.
 9. The system of claim 1, thesupport component including a crossing strut frame.
 10. The system ofclaim 9, wherein the crossing strut frame supports three centrallylocated leaflets and three outer leaflets.
 11. The system of claim 10,wherein the three centrally located leaflets together form a triangularshape when in a closed position.
 12. The system of claim 10, wherein thethree centrally located leaflet together form a generally circular shapewhen in a closed position.
 13. The system of claim 1, said leafletsincluding leaflets which include non-coaptive tips which extend belowthe native valve leaflets.
 14. The system of claim 1, said plurality ofleaflets comprising a pair of laterally arranged tri-leaflet valves. 15.The system of claim 1, the valve assembly configured to be suspendedwithin a coaptation line of a native valve.
 16. The system of claim 15,wherein said plurality of leaflets open and close in response tofunctioning of native leaflets.
 17. The system of claim 1, wherein saidplurality of leaflets are flap-like leaflets.
 18. The system of claim17, wherein certain of the flap-like leaflets are arranged at differentangles from other flap-like leaflets.
 19. The system of claim 1, whereinthe occluder component of the valve assembly comprises a tri-leafletconstruction in which said plurality of leaflets comprise threecentrally located leaflets together form a generally circular shape whenin a closed position.
 20. The system of claim 1, wherein the anchor isconfigured to seat along the native mitral valve annulus while thecommissural wire projections extend between the native mitral valveleaflets to maintain pre-existing native valve function.
 21. The systemof claim 20, wherein the commissural projections include loop structuresthat are shaped to anchor along the native heart tissue below the valveannulus of the native mitral valve.
 22. The system of claim 21, whereintwo of the commissural wire projections have the wire loop structuresthat are shaped to anchor proximate to trigones along the native hearttissue of the native mitral valve without interfering with thefunctionality of native mitral valve leaflets.
 23. The system of claim22, wherein at least a portion of the anchor is covered with ePTFEmaterial.
 24. The system of claim 23, wherein the valve assemblyincludes attachment components extending from the support component tomechanically fasten with the anchor.
 25. The system of claim 24, whereinthe attachment components of the valve assembly extend from the supportcomponent to mechanically fasten with a portion of the anchor above thenative mitral valve annulus.
 26. The system of claim 25, wherein thevalve assembly comprises a Nitinol frame and pericardial tissuematerial.
 27. The system of claim 26, wherein the valve assembly isconfigured to geometrically interlock with the anchor while preservingsubvalvular structural relationships.
 28. The system of claim 27,wherein the valve assembly is adjustable from a compressed state duringthe staged implantation to a deployed state in the heart.